Original on Original
In recent years, more and more third-party manufacturers have copied prosthetic components and claim compatibility with those of the original implant systems. At first glance the design of these non-original abutments seems to be equivalent to the corresponding original abutment. In fact, however, there are differences which cannot be discovered from the outside inspection, but can be visualized in a cross section of the implant-abutment system. Above all, the original prosthetic components show excellent long-term success and survival in clinical situations.
*Selected Publications Straumann® Original Components
Complication and Failure Rates with Implant-Supported Fixed Dental Prostheses and Single Crowns: A 10-Year Retrospective Study.
Wittneben JG, Buser D, Salvi GE, Bürgin W, Hicklin S, Brägger U. Clin Implant Dent Relat Res. 2013 Apr 2. PMID:23551688.
Screw loosening with interchangeable abutments in internally connected implants after cyclic loading.
Kim SK, Koak JY, Heo SJ, Taylor TD, Ryoo S, Lee SY. Int J Oral Maxillofac Implants. 2012 Jan-Feb;27(1):42-7. PMID: 22299077.
Implants with original and non-original abutment connections.
Gigandet M, Bigolin G, Faoro F, Bürgin W, Brägger U. Clin Implant Dent Relat Res. 2014 Apr;16(2):303-11. doi: 10.1111/j.1708-8208.2012.00479.x. Epub 2012 Jul 16. PMID: 22804848.
Long-term outcomes of dental implants with a titanium plasma-sprayed surface: a 20-year prospective case series study in partially edentulous patients.
Chappuis V, Buser R, Brägger U, Bornstein MM, Salvi GE, Buser D. Clin Implant Dent Relat Res. 2013 Dec;15(6):780-90. Epub 2013 Mar 18. PMID:23506385.
Exotic encounters with dental implants: managing.
Mattheos N, Janda MS. Aust Dent J. 2012 Jun;57(2):236-42. Epub 2012 Mar 16. PMID:22624768.
*Selected publications. No guarantee of completeness.